NewAge Silcon® Medical Grade Silicone Tubing
|New Age ||Special Application |Click to view a full size photo
|Application:||Silicone elastomer meets USP Class VI requirements.|
Reusable-will withstand repeated sterilization.
Nonreactive to body tissues and fluids.
May be used on peristaltic pumps.
Not recommended for implantable or in-body uses
| Tube:||Silicone Tubing|
|Temperature:||Brittle -100°F Maximum operating temperature 400°F|
individual, non-sterile, heat-sealed
|Couplings:||Thermobarb®, barbed fittings, Oetiker®, ear type clamps, Kwik Clamp&trade, nylon double bond hose clamps, Worm gear clamps|
|Options:||Size, Color, Hardness, and More...|
- Silicone elastomer meets USP Class VI requirements
- Manufactured under good manufacturing procedures in a
- Reusable - will withstand repeated sterilization
- Nonreactive to body tissues and fluids
- Translucent natural color for visual contact with the flow
- Odorless, tasteless, and inert
- All ingredients are non-toxic and FDA-sanctioned for use with
food contact surfaces
- May be used on peristaltic pumps
* Limited stock availability; lead times and minimums may apply — call JGB for details.
Sold by standard coil length only. Add length suffix to part number when ordering.
Example: 50 ft. of .030" I.D. x .065" O.D. tubing is part number 281 0304-50.
††Most common peristaltic pump sizes based on I.D. and wall requirements.
indicates the critical dimension for fittings application.
SILCON MEDICAL tubing is made under good
manufacturing procedures and conforms to the
biocompatibility requirements of USP Class VI.
SILCON MEDICAL’s surface properties resist
sticking and encrustation, and will not support
bacteria growth. The tubing is soft, pliable, and
contains no plasticizers which could leach out,
causing flow contamination or tube hardening.
SILCON MEDICAL is not recommended for
implantable or in-body uses.
Care is recommended in the selection and
application of fittings and clamps as sharp
barbed fittings or unlined metal clamps could
tear into the tubing wall.
If exposed to repeated steam sterilization or
long-term high temperature or pressure, silicone
will eventually relax and become gummy. It
should then be replaced.
Peroxide-cured SILCON MEDICAL tubing is
supplied in individual, non-sterile, heat-sealed
polybags and should be sterilized prior to use.
**Values listed are typical for the material used in
manufacture, except where noted, and are meant only as
a guide to aid in design. Field testing should be
performed to find the actual values for your application.
|Hardness, Shore A ±5
Tensile Strength, psi
Elongation at Break, %
Tear Resistance, Die B
Brittle Temperature, °F
Maximum Operating Temperature, °F
Comp. Set, 22 hrs. at 350°F, %